﻿ Summary of NTC Thermistor and Temperature Sensor_Industry knowledge_Nanjing Shiheng Electronics Co.,Ltd.

# 全民赢三张

News
Products
Honors
News Information
Industry knowledge

86-25-52121868

Location：Home > Industry knowledge > Text

### Summary of NTC Thermistor and Temperature Sensor

Published time：2011/6/28    Visitors：5829

Thermistor is a ceramic semiconducting element made from exorbitant oxides material。 It has the feature that the resistance changes according to the ambient temperature。 Namely, their resistance declines with the rising of ambient temperature at a determinated measuring power。 With this feature NTC thermistor and temperature sensor can be applied in the situation of temperature compensation, measurement and control and surge current protection。

Main Techno-Parameter of NTC Thermistor

1.Zero Power Resistance RT
At rated temperature, it is the resistance measured by the measuring power which causes the resistance change that can be ignored relative to the whole measuring error.

2.Rated Zero Power ResistanceR25
The rated resistance of thermistor which is the zero power resistance measured at 25℃ and signed on the thermistor.

3.B Value
B value is the thermal exponent of negative temperature coefficient thermistor, which is defined as the ratio of the difference between the napierian logarithms of zero power resistance at two temperatures to the difference between the two temperatures’ reciproal.

In the equation:
RT1- the zero power resistance at T1
RT2- the zero power resistance at T2

Unless the particular indication, B value is figured out from the zero power resistances at 25℃(298.15K) and 50℃(323.15K) and B value is not a rigorous constant in the range of operating temperature.

4.Temperature Coefficient of Zero Power ResistanceαT

where:
αT-the temperature coefficient of zero power resistance at T
RT-the zero power resistance at T
T-temperature
B- B value

5.Dissipation Coefficientδ
At rated ambient temperature, it is the ratio of consumption power change rate of thermistor to the change of the corresponding temperature, namely:

In the range of operating temperature, δ has a little change with the ambient。

6.Thermal Time Constantτ
At zero power, it is measured as time in seconds which needed for thermistor temperature change of 63.2% difference between initial and final thermistor temperature when the temperature breaks.
τis in direct ratio to thermal capacity C of thermistor and in inverse ratio to the dissipation coefficient, namely: